Bulletin # 6
REAL PEOPLE - WWW.THESOLARBIZ.COM - 888-826-0939
Before measuring the specific gravity, completely read Bulletin # 7 on Basic Battery Safety.
The most accurate and direct way to test the state of charge of a battery cell is to determine the specific gravity of the battery electrolyte. The higher the specific gravity of the electrolyte the higher the state of charge. The best way to truly monitor your system over it lifetime is to regularly take and record specific gravity readings.
hydrometers are not easy to use. Testing can be time consuming, there are
possibilities for error and safety must be considered. For these reasons we
present this bulletin.
Hydrometers come in
many sizes and shapes. We recommend a hydrometer with a float, contained in a
glass vessel with a rubber bulb to draw the acid into the tube. Stay away from
floating colored balls as the extra inaccuracy results in very subjective
testing. The hydrometer should give you a numeric reading directly from the
instrument. A good hydrometer is accurate to +/- 0.005 points so 1.265 could
read from 1.260-1.270. The instrument accuracy should be known.
As with all measuring equipment drawing conclusions from the results is not worth the effort if the equipment is not calibrated.
The correct way to check the calibration of a hydrometer is to test the hydrometer against a known master that is accurate to one more decimal point. These hydrometers are; however, very expensive ($100-150) and can be easily broken.
The simplest and cheapest way, when facilities exist, is to measure out a volume of acid and weigh it. A small graduated cylinder and an electronic balance are perfect. The specific gravity is then calculated as follows:
SG = Mass (g) / Volume (ml)
units must be used to convert to the specific gravity scale based on water. The
hydrometer is calibrated if it agrees with the sample within its manufacturing
Method of Use
Exact procedures is instrument dependent and this is given as a general procedure and assumes a hydrometer with glass float and body.
|Put on eye protection and rubber gloves. See bulletin 607, Basic Lead Acid Battery Safety Principals|
|It is recommended to disconnect the battery especially if on a high rate of charge / discharge.|
|Remove vent cap. Carefully insert the hydrometer into cell, not pushing down on the top of the plates.|
|Carefully draw liquid into the hydrometer and avoid "bumping" the hydrometer. Be careful the float is not flooded (too much liquid) or sticking to the sides of the glass tube.|
|Obtain a reading by looking directly at the float.|
|Repeat steps 3-5 to reconfirm reading.|
|RECORD the cell number and result.|
|If it is very warm or very cold correct the specific gravity for temperature. If the ambient temperature is fairly consistent and original gravities are taken when the batteries are put into service temperature correction is not as critical and only necessary if problems arise. Make sure electrolyte is not hot if just taken out of service. Let it reach room temperature.|
procedure is to number the cells starting with the positive cell and move from
cell to cell towards the negative terminal. If this is part of a preventive
maintenance program it is helpful to number the batteries.
SG of acid is temperature dependent. If the temperature is very cold or very hot this can lead to incorrect readings. To correct for temperature use the following equations the equations or below 70ºF subtract points (0.03 per 10ºF) and above 70ºF add points.
|Correction factor = (0.331 x T/ºF - 23) / 100 or 0.03 pts per 10 ºF|
|Correction factor = (0.595 x T/ºF - 12.5) / 100 or 0.03 pts per 5 ºC|
This is valid for 0-130ºF or -17.8- 54.4ºC
We have been working with solar energy & battery systems since 1969... Over Four Decades.
Forty Seven Years Experience!!
Our Solar Panel Prices are The Best on The Planet
· We are a full service company. We want a real person to consult with you because, we want to be sure that you don't waste your time and money and that you buy only what you need.
· We strongly recommend that you talk with us first about your solar power needs. Please use our expertise as a resource.
We have been working with solar energy & battery systems since 1969... Almost a Half Century of Experience. Our 47th Anniversary!!
If you think you find a better price anywhere, call us before you buy.
Call us Toll Free at: 1-888-826-0939
Central America: (Panamá) 507-836-5588
Other International: 1-575-539-2111
E-mail us at: email@example.com
PLEASE NOTE: Hours, Monday thru Friday 9:00 AM to 7:00 PM Central Time.
Except as otherwise noted, the text and graphics provided on our web site are copyrighted by The Solar Biz Inc. or its licensors. Material on this site is intended for your personal use only. The Solar Biz Inc. does, however, permit visitors to make a single copy of the information published on The Solar Biz Inc. web site for their personal, non-commercial use or use within the organization that employs them.
Pursuant to the Digital Millennium Copyright Act of 1998, use of this web-site’s unique copyright protected intellectual property or Registered Trademarks, without prior written permission, for any purpose including but not limited to advertising, promotion or sale of products or services, for business or private financial gain, any commercial advantage, or the enhancement of competing websites, or any other marketing media, constitutes an act of willful copyright infringement.
PLEASE NOTE: Such willful infringement provides for substantial financial penalties and potential criminal prosecution. Remedies for such violations include not only injunctive relief against the infringer, but also recovery of profits and damages suffered by the copyright owner. 17 U.S.C.A. § 504. In addition, such relief includes recovery of attorneys' fees and costs incurred in connection with litigating the matter where infringement is willful. For more information on THE DIGITAL MILLENNIUM COPYRIGHT ACT OF 1998 please visit: http://www.copyright.gov/legislation/dmca.pdf
Copyright © 1997- 2016 The Solar Biz ™. All Rights Reserved